Electrothermal Distilled Water Machine
( 5l, 10l )
applies to make distilled water for medicine and sanitation, chemical industry, science research companies and laboratories.
1. Made of high quality stainless steel through stamping and advanced welding. Beautiful shape and anti-corrosion.
2. Equipped observation glass in the middle of the evaporator to make us see the level of water and the situation of boiling.
3. Inner immersed heating tube is treated with copper, safety and high efficiency heating.
C. Main Technical Parameters
1. vapour valve 2. Backwater tube
3. condenser 4. disilled water out
5 water in vale 6. observation glass
7. evaporator 8. heating tube
9. electric parts 10. water out valve
11. mouth of water out 12. funnel
E. Operation Guid
1. Firstly, turn on the water out valve (10)
2. Then turn on the water in valve (5), let the water through the condenser (3), backwater tube (2), funnel (12), then into the evaporator (7). Turn off the water until we can see the water level from the observation glass (6).
3. Use the same voltage as the machine required, and also make the ground wire of jack to the ground safely.
4. After power indicator displays, the machine begins to heat. When the water begins to boil, let in some water again to the evaporator. Control the temperature of distilled water with the going up temperature of water (through controlling a given of amount water).
F . Attentions
1. This kind of machine is made of metal materials, so before using, you need clean it first, then evaporate it advanced for 2 hours, until the distilled water quality conforms with the standards. Every time before using, you need also evaporate it advanced for 30 minutes, then the distilled water could be used.
1. Advanced evaporating follow operation guide.
2. If the machine is out of water or the water level is lower than the heating tube, it will make the heating tube damage.
3. The bottom is installed in electric parts, so do not make it work in water or in damp environment.
4. If do not make the hydraulic pressure stable, it will in effect the distilled effect.
G. Installation and Maintenance
1. It should be installed in stable platform.
2.The source should be tap water.
3. The rubber tubes should be uniform with the machine’s tubes.
4. Water out tube and distilled water tube should not be to long. In order not to touch the distilled water, do not make the tube outlet in the water.
5. After using, let out all water and clean it. Make sure it dry and clean.
H. Storage and Movement
Make it in 20℃---40℃ shadow and dry environment.
Contact Name: Jully
Company Name: Xian HEB Biotechnology Co., Ltd
Street Address: No.460 Ke Chuang
Road Yan Ta District Xian , China
Member name: jully091
Member Since: 10 June 2010
Total Leads: 25 jully091 Import Export Business Leads
Business focus: Rotary Evaporator, Cylindrical Jacketed Reactor, Lcd Rotary Evaporator, Water Distiller, Bath Serise, Incubator, Oven, Centrifuge Series, Sterilizer Seris
Verify: Safe Import Export Tips